|Image file formats|
|Copyright||© 2004-2015 by Global Phasing Limited|
|All rights reserved.|
|This software is proprietary to and embodies the confidential technology of Global Phasing Limited (GPhL). Possession, use, duplication or dissemination of the software is authorised only pursuant to a valid written licence from GPhL.|
|Documentation||(2004-2015) Clemens Vonrhein, Claus Flensburg, Wlodek Paciorek & Gérard Bricogne|
Requests for support of additional detector types in the XDS modules of autoPROC should be accompanied by example images or datasets.
If the automatically determined description of the specific setup (direct beam, goniometer, detector) is inadequate, several parameters can be used to set them explicitly, e.g. using
autoPROC_XdsKeyword_INCIDENT_BEAM_DIRECTION="0.0 0.0 1.0" autoPROC_XdsKeyword_DIRECTION_OF_DETECTOR_XAXIS="1.0 0.0 0.0" autoPROC_XdsKeyword_DIRECTION_OF_DETECTOR_YAXIS="0.0 1.0 0.0" autoPROC_XdsKeyword_ROTATION_AXIS="0.0 -1.0 0.0"As you can see, any XDS input parameter can be set directly using the autoPROC_XdsKeyword_* syntax. We also provide a wiki page with beamline/instrument-specific settings, that we attempt to keep up-to-date as much as possible. Please contact us if a particular instrument is not listed there or if you had to use non-standard settings on a particular detector/instrument.
Note : it is recommended to use the latest version of XDS (at the time of writing: March 1, 2015)